7-16. LASER POWER METER (INTERNAL RADIOMETER) 2A1A7
2. CORRECTIVE ACTION - INTRAD (cont)
(d) Check the following cable assemblies and repair as required:
If any cable was repaired, perform INTRAD test. If INTRAD test passes,
no further action required.
If INTRAD test fails and CALCST, PLLBUS,
MTRXSW, ADST, DIGITST, FPSST, and PPSST tests have been performed,
repair is beyond this level of maintenance.
Next higher level of
If no cables were repaired, repair is beyond this level of maintenance.
Next higher level of maintenance required.