5-7. BASE ASSEMBLY ADJUSTMENT (cont)
If measurements are not equal, per-
form step 9.
If measurements are
equal, go to step 10.
When adjustment of isolator No. 3
valve adjusting arm is made, both
No. 3 and No. 4 isolators are
Loosen nut (10) on appropriate valve
adjusting arm (11) and turn height
adjusting screw (12) counterclockwise
to decrease gap and clockwise to
Tighten nut (10) when
desired gap is achieved.
10. Apply sealing compound primer
to screw (12) and nut (10). Allow to
11. Apply sealing compound to screw (12)
and nut (10).
12. Close door on nightside test bench.
END OF TASK
5-8. CABLE ASSEMBLY AND CONNECTOR REPAIR/REPLACEMENT
Repair or replace cables and connectors using standard maintenance practices.
For electro-optical test bench set interconnection and cable assembly wire list data,
refer to chapter 3.
The cable assemblies are listed in reference designator sequence
(2A2W1, 2A2W2, etc.).